This product is no longer in stock
Primary Filter : 4.75" x 23.75"
115 volt / single phase / 50-60 hertz
5 amp minimum requirement
High Pressure Direct Drive Radial Blade Blower
1/5 Horse Power 3 speed motor
(wired for high speed only)
550 cfm. Air Flow
Do not pour liquid waste into this unit. It has
been designed to handle used animal bedding
material and the associated moisture in that
Introduction of liquid into the unit will create a
shock hazard for operators and personnel in the
area. In addition, void any associated warranty
of the motor and filter unit.
in accordance with this manual may damage the
unit which Mopec will not be responsible.
HOW TO CHANGE FILTERS UNITS
Replaceable filter elements, both primary and
HEPA, are to be replaced by the customer at
Unit not should be operated with panels removed or door open.
How to change filter units.
Upon removal of the perforated plate, the primary filter
can be lifted out of the holder and replaced by new media.
The media should be firmly inserted into the holder frame so
that it seats to the bottom of the frame.
In order to replace the HEPA filter, the power cord
needs to be unplugged from the outlet. To provide additional
access space the louvers on the back panel need to be
placed in the closed position. This is accomplished by
pushing the two levers down in the bottom corners of the
Remove the left and right side service panels by
pushing in the latch, tilting the door outward via the latch.
Pick the door up and off the locator pins using two hands on
each side of the door.
Through the larger opening, slide the HEPA unit to the
right until the foam seal clears the door panel opening. At
this point the HEPA filter can be removed.
Upon turning the unit on, the pressure gauge will read
0.38 As depicted in the picture below.
This is a differential pressure gauge. A reading of “0”
indicates a 100% blockage at the primary filter.
The primary filter should be cleaned daily to ensure that
the unit is operating at maximum efficiency
Should cleaning of the primary filter not produce a 0.38
or greater reading, replacement of the primary is necessary.
Through use a slow steady deflection to the right will
occur, this indicates the HEPA filter is working. As the
gauge approaches 2.0 inches, the HEPA filter needs to be
Before changing the HEPA filter the primary filter
should be replaced; or inspected, cleared of debris and
CLEANING AND MAINTENANCE
All stainless steel surfaces can be cleaned with soap and water, which will
The stainless steel surfaces can be disinfected with a non caustic disinfectant.
During the cleaning we suggest wiping the surface in the same direction as the
satin finish which will help elevate dirt from the grain finish.
Scratches can be removed simply by utilizing a "non-metallic" abrasive pad and
rubbing in the same direction as the satin finish.
Since most abrasive pads vary from one supply to another, we suggest rubbing
the entire surface to blend the scratch and blend the balance of the surface.
Do not use a bleach solution.
To clean external and internal surfaces use
cleaning agents for stainless steel only. The
agent should be applied to a foam cleaning pad
prior to contacting the stainless steel in order to
contain the agent in use.